New insulin pump from Tandem, the T:Slim touch screen insulin pump.
Please note: the exchange, sale or giveaway of items between members that require a prescription from a licensed practitioner, including insulin pumps and pump supplies, is not allowed on TuDiabetes.
We encourage you to donate supplies to non-profits such as the Charles Ray III Diabetes Association and Insulin for Life, which accept insulin pumps and pump supplies (as well as other diabetes-related prescription items).
You can also approach your physician's office or local medical groups to discuss donating them to those in need of assistance in your area.
Latest Activity: on Saturday
After having owned my T:slim for almost two months (love it love it love it), I was wondering if anyone is successfully using Apidra?In the beginning there was an issue with Apidra constantly causing occlusions, but it was also found that some of…Continue
So I spent 7 hours in ER today. Mild DKA. Changed my infusion first thing this morning and 2 hours later BG was almost 400. Changed the set again and notice the first set had a 90 degree bend at the end, so I thought that was the problem. My BG…Continue
I've had my pump for a year now, and the past couple of months I've noticed that once I get down to about 40-50 units remaining, my pump starts having some issues... It will give me at least a couple of occlusion alerts every time that I bolus, and…Continue
I have had my T:Slim for about 3 weeks and love it. But in trying to use T:Connect all I can get is the Dashboard. If I click on Reports a lll I get is the tabs at the top of the page for Trends etc. I am using a Mac Desktop with the latest version…Continue
Has anyone else run into this alarm on the t:slim?I received it not after a flight, but rather after a hot day of bike riding. I wear my t:slim against my hip when I ride and I have been doing this for 6 months. But this past weekend after coming…Continue
Hello everyone, I am new here and have an appointment with my CDE on Thursday morning so I can try the T-slim. If all goes well, I will officially switch following the trial! I wanted to get your feedback on a conversation I had with a Medtronic…Continue
When I was trained on my t:slim it was recommended that I charge the battery a little every day. But I found that not to be too practical for me. So now I just let the battery drain and then give it a full charge. The lowest battery % i've gone down…Continue
Hey everyone! I'm new to the site, so bear with me if something like this has been asked before! I've been on the Animas Ping pump for just under 5 years now, and I absolutely love it. But since I've been introduced to the T:slim, I can't stop…Continue
So I had an Occlusion alarm. I thought it was odd since I changed the gear the day before. I was in the middle of a bolus. I ignored the alarm and then gave myself the rest of the bolus. No alarm this time!Has this happened to anybody? Is anybody…Continue
Always and I do mean ALWAYS pack extra supplies. Today could have turned out so much worse had I not had an extra Rx on file for Humalog Quick Pens I would have had to wait nearly 30 minutes for any insulin at all. I was changing the cartridge on…Continue